August 1, 2002Greenlee Textron Has Recalled Electrical
- Greenlee Textron, Inc. has recalled about 650 electrical testing meters.
The meters are used to measure voltage and current of electric-powered
equipment. An incompatible grommet, located in the battery compartment to
protect internal wires, can cause the meter to provide inaccurate voltage
and current readings. A meter that inaccurately indicates zero voltage or
current creates the potential for electric shock or an electrocution hazard
from the equipment being tested.
CPSC and Greenlee have not received any reports of injuries or property
damage. This recall is being conducted to prevent the possibility of
This recall involves certain CM-700 and CM-750 electrical meters. The 8-inch
meters are dark green with bright yellow features. The model number and the
words "Greenlee Test Instruments" are printed across the front of the meter.
A silver plate on the back of the meter displays the serial number (S/N).
Serial numbers included in the recall range from 0203540001 through
0203540650. The electrical meters were made in Taiwan.
Home Depot (model CM-750 only) and electrical supply stores sold the meters
nationwide during April 2002. The CM-700 model sold for about $100 and the
CM-750 for about $140.
Consumers should stop using the recalled meters immediately and contact
Greenlee to receive a replacement meter. For more information, consumers can
contact Greenlee toll-free at (800) 435-0786 between 8 a.m. to 5 p.m. CT
Monday through Friday.
CM-700 and CM-750 meters outside the above serial number range are not
included in this recall.